wouldn't a thermometer in the reservoir validate the functionality of
the heat exchanger? also the laser should shut down on an over temp
prior to any damage? and what serious damage will it cause? is the
detailed literature too long to post on the m/l?
>From email@example.com Mon Mar 08 14:20:40 1999
>Received: from [22.214.171.124] by hotmail.com (1.2) with SMTP id
MHotMailB8AD982E460CD1017077C1A642016C390; Mon Mar 08 14:20:40 1999
>Received: from major by bart.lpt.fi with local (Exim 1.90 #2)
> for firstname.lastname@example.org
> id 10K7wk-0007PZ-00; Mon, 8 Mar 1999 21:52:22 +0000
>Received: from gull.prod.itd.earthlink.net ([126.96.36.199])
> by bart.lpt.fi with esmtp (Exim 1.90 #2)
> for email@example.com
> id 10K7wj-0002Tf-00; Mon, 8 Mar 1999 23:52:21 +0200
>Received: from LOCALNAME (pool189-cvx.ds38-ca-us.dialup.earthlink.net
> by gull.prod.itd.earthlink.net (8.8.7/8.8.5) with SMTP id NAA11265
> for <firstname.lastname@example.org>; Mon, 8 Mar 1999 13:52:14 -0800 (PST)
>Date: Mon, 08 Mar 1999 14:55:54 -0800
>From: ERIC KING <email@example.com>
>X-Mailer: Mozilla 3.0C-NSCP (Win16; U)
>Subject: SLA-500, heat exchangers
>Content-Type: text/plain; charset=us-ascii
>Please check your heat exchanger.
>It has come to our attention that there are several problems
>concerning the Laser Pure heat exchangers on the SLA-500 machines:
>1) The Coherent "Laser Pure 20" heat exchangers out in the field,
> are not the correct model, and may cause severe damage to your
> laser. They were not intended to be used with this model laser,
> and we can supply detailed literature upon your request.
>2) We have also seen "Laser Pure 40" models with the wrong
> internal components. Which also may cause laser problems.
> To check for the correct component, remove the top lid,
> and remove the side panel from the LP-40 unit.
> (the side that has all the water fittings)
> Locate the stainless steel heat transfer unit.
> The unit should have the model # CB26-90H,
> if it does not, please contact us.
>We have only seen these problems, on systems with dates prior to
>If you have questions, or need more info, please contact us:
>(805) 529-5864 tel
> 6621 fax
>For more information about the rp-ml, see http://ltk.hut.fi/rp-ml/
Get Your Private, Free Email at http://www.hotmail.com
For more information about the rp-ml, see http://ltk.hut.fi/rp-ml/
This archive was generated by hypermail 2.1.2 : Tue Jun 05 2001 - 22:51:10 EEST